# Two Wattmeter Method of Power Measurement

**Two Wattmeter** Method can be employed to measure the power in a 3 phase, three wire star or delta connected the balanced or unbalanced load. In Two wattmeter method the current coils of the wattmeter are connected with any two lines, say R and Y and the potential coil of each wattmeter is joined on the same line, the third line i.e. B as shown below in figure (A).

The total instantaneous power absorbed by the three loads Z_{1}, Z_{2} and Z_{3}, are equal to the sum of the powers measured by the Two wattmeters, W_{1 }and W_{2}.

**Contents:**

## Measurement of Power by Two Wattmeter Method in Star Connection

Considering the above figure (A) in which Two Wattmeter W_{1} and W_{2} are connected, the instantaneous current through the current coil of Wattmeter, W_{1} is given by the equation shown below.

Instantaneous potential difference across the potential coil of Wattmeter, W_{1} is given as

Instantaneous power measured by the Wattmeter, W_{1} is

The instantaneous current through the current coil of Wattmeter, W_{2} is given by the equation

Instantaneous potential difference across the potential coil of Wattmeter, W_{2} is given as

Instantaneous power measured by the Wattmeter, W_{2} is

Therefore, the Total Power Measured by the Two Wattmeters W_{1} and W_{2} will be obtained by adding the equation (1) and (2).

Where P – the total power absorbed in the three loads at any instant.

## Measurement of Power by Two Wattmeter Method in Delta Connection

Considering the delta connected circuit shown in the figure below.

The instantaneous current through the coil of the Wattmeter, W_{1} is given by the equation

Instantaneous Power measured by the Wattmeter, W_{1 }will be

Therefore, the instantaneous power measured by the Wattmeter, W_{1} will be given as

The instantaneous current through the current coil of the Wattmeter, W_{2} is given as

The instantaneous potential difference across the potential coil of Wattmeter, W_{2} is

Therefore, the instantaneous power measured by Wattmeter, W_{2} will be

Hence, to obtain the total power measured by the Two Wattmeter the two equations, i.e. equation (3) and (4) has to be added.

Where P is the total power absorbed in the three loads at any instant.

The power measured by the Two Wattmeter at any instant is the instantaneous power absorbed by the three loads connected in three phases. In fact, this power is the average power drawn by the load since the Wattmeter reads the average power because of the inertia of their moving system.

## 12 Comments

What type of process we can run a single phase motor in reverse mode?

The motor can run in reverse mode by reversing the line connections of either the main winding or the starting winding.

Advantage of using two wattmeter method

1. Two wattmeter method measures the three phase power.

2. Useful for both the balanced and unbalanced loads.

Thanks.

1.use in balanced and unbalanced load

2.no need to connect neutral wire

Thanks.

Great Article.

Awesome 🙂 🙂

Very Informative. Please add more information regarding different types of connections made with watt meter in three phase systems…..

Thanks:)

Is there any serious errors in measurement of power through watt metre of 1 and 2 it there any how to minimise it